Have you forgotten your password?
No problem, you can set up a new password here.
When you click on 'Request Password', you will be sent an e-mail with instructions on how to set up a new password.
Please be sure to input the exact address that you used when you registered with Praxisdienst. Otherwise, your e-mail address cannot be found by the server.
If you are still having problems logging in, despite inputting the correct e-mail address, and you did not receive the "password forgotten" e-mail, please contact us by Email at the following address: email@example.com
With Reflotron® Clean + Check, the Reflotron® blood analyser can be easily cleaned and tested. The Clean + Check set contains 16 cleaning wipes and 15 test strips. The individually-packed cleaning wipes are used for cleaning the measuring system according to the instructions provided. The cleaning should be done when the Reflotron® is displaying the message "Measure Clean and Check". This occurs after 7 days or 100 measurements. After cleaning, the Reflotron® blood analyser is turned back on and the Clean + Check test strip inserted.
The Reflotron® Clean + Check test strip checks the optical system. The measurement is OK when the sample values lie within the reference range indicated on the packaging.